The IFC signal converter is designed to measure the flow velocity, IFC signal converter can be combined with any KROHNE flow. The Krohne IFC signal converter will need to be setup to output pulses from the flow meter. This guide will outline the procedure on the IFC signal. Ifc • Read online or download PDF • KROHNE IFC Converter Quickstart EN User Manual.
|Country:||Papua New Guinea|
|Published (Last):||22 September 2015|
|PDF File Size:||14.52 Mb|
|ePub File Size:||7.65 Mb|
|Price:||Free* [*Free Regsitration Required]|
Field frequency is not reaching steady state. Or electrodes completely insulated e. Measuring pipe not filled, function. Display functions Operating menu Setting Application test or operator action necessary. Open the catalog to page 4.
The difference lies in. This email is already in use. Current not correct, mA output cable has. F backup 2 settings. The difference lies in whether the measured value is also set to zero upon the detection of an empty pipe. You have one of those, right? F active settings Error during the CRC check of the active settings. F backup 2 settings F wiring A Mwnual current at iffc current input is less than 0.
This latest member of the converter family also has a large number of fully-developed functions: Measured values are still supplied, but they are always too low. Measuring pipe not filled, function dependent on Fct.
Complain wrong Brand wrong Model non readable.
Error during the CRC check of the factory. Wir freuen uns, dass Sie ein Mitglied geworden sind!
KROHNE IFC signal converter manual — Visaya eShop
The current at the current input is less than. F over range A The pulse rate or the corresponding measured value is limited by a filter setting. Storage temperature Pressure Medium Refer to technical data for the measuring sensor.
No message if empty pipe. Diese Information hilft uns um eine professionelle Community manuxl Visaya aufzubauen.
Unsere Lieferzeiten während der Feiertage
F flow exceeding limit. The electronics can be used in all housing versions without having to be reconfigurated. If the measuring point is difficult to access or the ambient conditions prevent the use of the compact variant, the signal converter is available If this limit occurs sporadically in processes with air pockets, solid ufc or low.
Save active settings in backup 1 or 2.
F over range A The current or the corresponding measured value is limited by a filter setting. No middle ground here! It is a good idea to protect the converter from external heat sources such mqnual direct sunlight as higher temperatures reduce the life cycle of all electronic components.
This magnetic field is generated by a current, flowing through a orohne of field coils. F application error Application-dependent fault, but device is OK. Measuring range exceeded, filter setting.
Please check your inbox. F factory settings Error during the CRC check of the factory manaul. Ambient temperature Depending on the version and combination of outputs. If this limit occurs sporadically in processes with air pockets, solid contents or low conductivity, then either the limit has maanual be increased or a pulse filter used so as to quell the error messages and also reduce the measurement errors. The two empty pipe messages cannot appear at the same time.
Open the catalog to page 5.
Current not correct, mA output cable has open circuit or load too high. F open circuit A Load on current output A too high, effective current too low.
Or the demanded pulse rate is too high.
IFC – KROHNE Messtechnik – PDF Catalogs | Technical Documentation | Brochure
I so hate finding manuals on some of these vendor sites. Could you give me the right link to download?. F field frequency too high Field frequency is not reaching steady state, a measured flow value is still being supplied but may have errors.