YOKOGAWA EJAA, EJAA Absolute and Gauge Pressure Transmitters EJAA/HAC general specifications · EJAA and EJAA User’s Manual. Yokogawa Dpharp vigilantplant EJAA Pdf User Manuals. View online or download Yokogawa Dpharp vigilantplant EJAA User Manual. Jun 9, Regarding This Manual. • This manual should be provided to the end user. • The contents of this manual are subject to change without prior.
|Published (Last):||3 July 2010|
|PDF File Size:||8.91 Mb|
|ePub File Size:||15.32 Mb|
|Price:||Free* [*Free Regsitration Required]|
Looking for more information on our people, technology and solutions?
Is there any pressure leak? ESC key twice to F Yokogawa Electric Corporation, a leading company in industrial automation IA and technology solutions, recently announced the completion of its storey building in Singapore.
Use a slotted screwdriver to turn the zeroadjustment screw. Press the OK key after confirmation.
EJAA In-Line Mount Gauge Pressure Transmitter | Yokogawa Electric Corporation
General-use Type And Flameproof Type This subsection describes the procedure for replacing an integral indicator. Outputs high range limit value or low range limit value.
To prevent such problems, the process pressure taps must be angled as shown in Figure 5. The new facility will play a crucial role in Yokogawa’s vision to be the top global IA player by Rotating Transmitter Section Refer to Section 6.
Leave this resistor connected at least one second to discharge any static potential. If the error code still exists, replace the capsule assembly. Signal without carrier [Baud rate] bps [Communication signal] host to device: Save an inspection date of January 30, Liquid and Gas Group of Fluid: Use heat insulation or eja530 lagging to keep temperature within range. See Chapter 8 for BT operation.
Model Kanual Suffix Codes Power Supply Wiring Connection 6.
Alphanumeric Keys And Shift Keys Even if the BT power supply is turned off or the communication cable is disconnected during test output, it is eja5330a for approximately 10 minutes. If a pressure generator is used: This transmitter is equipped with a self-diagnostic function which will be useful in troubleshooting; see Section 8. Also make sure before storing that the transmitter assemblies are securely mounted.
Checking With Integral Indicator Mounting position effects or zero shifts caused by static pressure are typically compensated by a zero adjustment. When a zero point adjustment has been made, do not turn off the transmitter less than 30 seconds after adjustment. Follow the instructions in section 9. Switching language of messages Japanese or English 4.
Calibration Instruments Selection Drain plug Flexible metal conduit Receiver instrument F Also check that all transmitter mounting hardware shown in Figure 2. Before connecting the transmitter to the process, ejw530a the eja53a0 installation location, the process piping layout, and the characteristics of the process fluid corrosiveness, toxicity, flammability, etc.
Thereafter, always use the transceiver outside the area affected by noise. August Yokogawa’s new facility in Singapore will play vital role in its regional and international growth.
EJA530A In-Line Mount Gauge Pressure Transmitter (DISCONTINUED)
NOTE Whenever shutting down the eua530a for a long period, remove manuual process fluid from the transmitter pressure-detector section. Note that the amount of adjustment to the zero point changes according to the speed at which the screw is turned. Accurately adjust the output current to 4 mA DC or other target screw output value using an ammeter that P. In such case, adjust the transmitter output signal to the actual measured value obtained from a high-accuracy pressure measuring instrument.
Yokogawa EJA115 User Manual
Press ejq530a to enter the setting. When the shrouding bolt is driven clockwise by an Allen wrench, it is going in and cover lock is released, and then the cover can be opened by hand. If the normal pressure is regained, the Er. Hysteresis of cut point: Abnormalities appear in measurement. Emc Conformity Standards